Q: Which items should be inspected and replaced during the preventive maintenace?
A: Remove the product from service before you perform the preventive maintenance inspection. Check all items listed during
annual preventive maintenance for all Stryker Medical products. You may need to perform preventive maintenance checks
more often based on your level of product usage. Service only by qualified service technician.
Note: Complete functional checks as indicated in the maintenance manual.
Inspect all of the following items:
_____All fasteners secure
_____Power cord and plug for fraying
_____ Condition of covers and push handle for damage
_____ Hose ports are operational
_____ Ground chain attached
_____ LCD is not cracked
_____ LCD operational
_____ Touch screen operational
_____ Wheels for smooth operation
_____ Rear caster wheels for free swivel action
_____ Both rear wheels lock secure when applied
_____ Front and rear wheels are not loose or wobbly
_____ Battery backup, functional check
_____ Alarm system - visual and audible, functional check
_____ High temp cutout checks
_____ Water temperature and flow verification, functional check
_____ Probe resistance, functional check
_____ Clear RFU codes
_____ Ground impedance not more than 100mΩ (millohms)
_____ Current leakage not more than 300 (microamps)
_____ Integrity of all clamps and clamped joints located in the air elimination circuit.
Replace the following on an annual basis:
_____ 9 V battery
_____ Condenser inlet filter
_____ Air eliminator hose
_____ Inlet filter